##TITLE= Audit trail, TopSpin 4.3.0
##JCAMPDX= 5.01
##ORIGIN= Bruker BioSpin GmbH
##OWNER= nmr
$$ /opt/nmrdata/user/NMR/data/arthur/240520_HPMCS2/54/pdata/1/auditp.txt
##AUDIT TRAIL=  $$ (NUMBER, WHEN, WHO, WHERE, PROCESS, VERSION, WHAT)
(   1,<2024-05-22T16:05:15.789+0200>,<nmr>,<pink>,<cprserver>,<TopSpin 4.3.0>,
      <new/edc PROCNO=1>)
(   2,<2024-05-22T16:12:55.264+0200>,<nmr>,<pink>,<proc1d>,<TopSpin 4.3.0>,
      <Start of raw data processing
       efp LB = 100 FT_mod = 6 PKNL = 1 PHC0 = -68.917 PHC1 = -117.702 SI = 4K 
       data hash MD5: 4K
       E9 AA B3 97 C8 2D F9 F1 42 BB 7E 35 6C 9D 71 4C>)
(   3,<2024-05-22T16:13:20.553+0200>,<nmr>,<pink>,<proc1d>,<TopSpin 4.3.0>,
      <pk fgphup PHC0 = -14.50752 PHC1 = 30.8 
       data hash MD5: 4K
       BD E6 26 D5 FB 07 69 D0 3F E6 DC 8C 7D 93 70 F9>)
(   4,<2024-05-22T16:24:18.312+0200>,<nmr>,<pink>,<audit>,<TopSpin 4.3.0>,
      <user comment:
       TITLE=MW ON
       Probe: 3.2 mm DNP HCN mode, AVNeo console
       Sample: sample 7 HPMC-AS (MF) repeat
       Saphire rotor
       MAS = 8 kHz, BP = 2656 mB, DP = 1078 mB
       Tsample = 124.4 K, flow = 2000 l/h, Heater 0%, Tbearing = 110.2 K, Heater 5%, Tdrive = 114.7 K, Heater 5%
       I = 100 mA
       Tcavity = 22.5 C
       13C CP echo no filter
       
       1H pulses and SPINAL dec 100 kHz at 67 W, o2 1H offset = -64442 Hz, pcpd2 = 4.8 us
       1H CP 100 kHz matched to 13C at 60 W (71 kHz), ramp 70.100.100 on 1H
       1 ms CP
       L0 = 3
       At temp in file 53TITLE_END
       data hash MD5: 4K
       BD E6 26 D5 FB 07 69 D0 3F E6 DC 8C 7D 93 70 F9>)
(   5,<2024-05-22T16:24:37.202+0200>,<nmr>,<pink>,<audit>,<TopSpin 4.3.0>,
      <user comment:
       TITLE=MW ON
       Probe: 3.2 mm DNP HCN mode, AVNeo console
       Sample: sample 7 HPMC-AS (MF) repeat
       Saphire rotor
       MAS = 8 kHz, BP = 2647 mB, DP = 1073 mB
       Tsample = 124.2 K, flow = 2000 l/h, Heater 0%, Tbearing = 109.8 K, Heater 5%, Tdrive = 114.7 K, Heater 5%
       I = 100 mA
       Tcavity = 22.5 C
       13C CP echo no filter
       
       1H pulses and SPINAL dec 100 kHz at 67 W, o2 1H offset = -64442 Hz, pcpd2 = 4.8 us
       1H CP 100 kHz matched to 13C at 60 W (71 kHz), ramp 70.100.100 on 1H
       1 ms CP
       L0 = 3
       At temp in file 53TITLE_END
       data hash MD5: 4K
       BD E6 26 D5 FB 07 69 D0 3F E6 DC 8C 7D 93 70 F9>)
(   6,<2024-05-23T19:14:45.015+0200>,<nmr>,<pink>,<proc1d>,<TopSpin 4.3.0>,
      <pk fgphup PHC0 = 5.201465 PHC1 = -6 
       data hash MD5: 4K
       66 B3 31 0F 85 6A 86 6C 75 FC C5 3C 6F 6E 61 DE>)
$$ /Users/ronancosquer/Documents/NMR data/PhD/240520_HPMCS2/54/pdata/1/auditp.txt
(   7,<2024-06-06T16:31:13.601+0100>,<ronancosquer>,<dh127142.ch.liv.ac.uk>,<proc1d>,<TopSpin 4.4.0>,
      <Start of raw data processing
       em LB = 100 SI = 4K 
       data hash MD5: 4K
       C2 C6 3B A8 0D 7E 1E 3E 6A 01 94 C5 48 95 C8 4F>)
(   8,<2024-06-06T16:31:13.613+0100>,<ronancosquer>,<dh127142.ch.liv.ac.uk>,<proc1d>,<TopSpin 4.4.0>,
      <ft FT_mod = 6 PKNL = 1 SI = 4K 
       data hash MD5: 4K
       B6 91 85 E9 A8 30 94 A0 C9 48 72 1E CC AF 97 31>)
(   9,<2024-06-06T16:31:13.624+0100>,<ronancosquer>,<dh127142.ch.liv.ac.uk>,<proc1d>,<TopSpin 4.4.0>,
      <apk 
       data hash MD5: 4K
       B4 1C 97 DE 15 0D FD D7 3D 26 85 A6 D9 2D 7B B8>)
(  10,<2024-06-06T16:31:13.634+0100>,<ronancosquer>,<dh127142.ch.liv.ac.uk>,<proc1d>,<TopSpin 4.4.0>,
      <abs n ABSG = 5 
       data hash MD5: 4K
       F2 FD 49 F6 E1 F2 87 57 EF 28 60 E1 58 A6 A0 FE>)
##END=

$$ hash MD5
$$ C8 36 A3 EA D4 75 D3 8E EE 00 D2 D4 79 5E 0F 13
